What is the calibration period of IBP sensors and meters?
The calibration period is one year. We recommend to double check the
conductivity once per month with a
conductivity reference solution. If biological
material grows inside the cell, it could interfere the accuracy
know that competing companies do not recommend a monthly test of the
some even offer this as a "feature" of their meters.
From our more than 25 years experience in developing
dialysis meters we strongly recommend monthly testing as a safeguard precaution.